Data for: Validation of ABI 3500xL Genetic Analyzer after decommissioned and recommissioned at new permises
Description of this data
This validation was aimed to demonstrate that there is no effect on the Applied Biosystems® 3500xL Genetic Analyzer (Thermo Fisher Scientific®, UK) after decommissioned and recommissioned at new premises. Same sample sets were tested on the ABI 3500xL Genetic Analyzer at both old and new premises using current accredited procedures. The results showed genotyping concordance which means using the ABI 3500xL Genetic Analyzer at the new premises provides no risk regarding the accuracy, precision and reproducibility of the results and also showed a very good performance on sensitivity, specificity, repeatability and good profile quality based on the assessment of locus peak balance and peak height.
Experiment data files
This data is associated with the following publication:
Cite this dataset
Iyavoo, Sasitaran; Haizel, Thomas; Draus-Barini, Jolanta (2017), “Data for: Validation of ABI 3500xL Genetic Analyzer after decommissioned and recommissioned at new permises”, Mendeley Data, v1 http://dx.doi.org/10.17632/h3bh292rcg.1
The files associated with this dataset are licensed under a Creative Commons Attribution 4.0 International licence.